ASTM D Standard Test Methods for In-Place Density and Water Content of Soil and Soil-Aggregate by Nuclear Methods (Shallow Depth). depth, etc.) Verify proper rebar sizes and placements. Verify proper reinforcing overlaps. Verify proper clearance and that chairs are. ASTM D Uploaded by Xavior. Standard Test Method for In-Place Density and Water Content of Soil and Soil- Aggregate by Nuclear Methods (Shallow.
|Genre:||Health and Food|
|Published (Last):||4 December 2013|
|PDF File Size:||13.46 Mb|
|ePub File Size:||10.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Soil testing Field Services: Users of this standard are cautioned that compliance with Practice D does not in itself ensure reliable results.
ASTM D6938 – 17a
Measurements for water moisture content are taken at the surface in backscatter mode regardless of the mode being used for density. It is beyond the scope of this standard to consider significant digits used in analysis methods for engineering design.
The water mass per unit volume is determined by the thermalizing or slowing asmt fast neutrons by hydrogen, a component of water. The direct transmission is the preferred method.
The values in inch-pound units ft — lb units are provided for information only. For more information, contact: The quality of the result produced by aatm standard test method is dependent on the competence of the personnel performing it, and the suitability of the equipment and facilities used.
For Individuals with needing special assistance for disabilities or limited English proficiency Please contact our Title VI specialist at In addition, they are representative of the significant digits that should generally be retained.
The nondestructive nature allows repetitive measurements at a single test location and statistical analysis of the results. When divided by the density of water and then multiplied bythe water mass per unit volume is equivalent d69388 the volumetric water content. This places the gamma and neutron sources and the detectors in the same plane.
Historical Version s – view previous versions of zstm. The total or wet density of soil and soil-aggregate is measured by the attenuation of gamma radiation where, in direct transmission, the source is placed at a known depth up to mm 12 in.
The density of the test sample in mass per unit volume is calculated by comparing the detected rate of gamma radiation with previously established calibration data. Request Quote Client Login.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the asttm.
ASTM D Standard
Link to Active This link will always route to the current Active version of the standard. Soil testing Back to Results Print.
Most gauges display this value directly. The neutron source and the thermal neutron detector are both located at the surface of the material being tested.
This d6398 method describes the procedures for measuring in-place density and moisture of soil and soil-aggregate by use of asgm equipment. Measurements for water moisture ashm are taken at the surface in backscatter mode regardless of the mode being used for density. Reliable results depend on many factors; Practice D provides a means of evaluating some of those factors.
Most gauges display this value directly. The water content most prevalent in engineering and construction activities is known as the gravimetric water content, w, and is the ratio of the mass of the water in pore spaces to the total mass of solids, expressed as a percentage.
The dry density of the test sample is obtained by subtracting the water mass per unit volume from the test sample wet density Section It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.